How should I exchange chambers & diodes during scanning?
Before disconnecting the chambers from the CCU (Common Control Unit), it is recommended to turn off the voltage to the channel(s).
Within the Controller panel choose the Electrometer panel tab. Choose the Field tab, and click the "Off" option under Output. Repeat the process for the Reference channel.
If more scans are to be conducted, connect the appropriate chambers back to the CCU.
***Ensure the correct detectors are selected in the Queue Setup section prior to applying voltage to either channel as damage to the detector could occur***
Turn the output back on for the new detectors and set the appropriate voltages if needed.