There is a water level detector in SMARTSCAN Water Reservoir, how to use it?
Please follow below steps to use the water level detector:
Connect the water hose to the water tank by pushing the two counterparts of the quick coupler together.
Connect the pump control connector (the blue cable ) to the Water Reservoir connector in the phantom connection box.
Connect the cable with the red connector to the water level detector cable.
Plug the power cord to the mains supply.
Install the water level detector in the phantom with the large universal holder.
Use the CCU hand control to move the detector to the desired water level.
For filling, mount the detector horizontally (below left); for emptying the water, mount the detector vertically (below right) and move it to the lowest position.
Press the Fill /Drain button on the water reservoir hand control. When water level is reached, the pump stops and the valve closes automatically.