Can a Dosimax plus detector (DEDX, RQX, RQA or RQM) be damaged if the wrong menu selection is made?
There will be no damage if your Dosimax plus is equipped with several detectors, e.g. DEDX and RQX, and you physically connect the DEDX but select RQX .
This also applies to any combination of the detectors DEDX, RQX, RQA and RQM.
An incorrect selection combination (e.g. DEDX connected and RQX selected) leads to an incorrect measurement value display.
Various calibration factors are stored in the Dosimax plus base unit for each available detector.
For this reason, the detector that is actually connected to the device must always be selected in the menu.