What can I do when the MatriXX measurement is not uniform under square field?
When the MatriXX measurement is not uniform, below suggestions may help you:
- Warm-up the device for at least 15min. If the MatriXX is not been used for a long time, a longer warm-up time (several hours) may help improving device performance.
- Pre-irradiate the complete MatriXX chamber field before measurement.
- If the above methods don't help, we suggest performing a user uniformity correction.