I forget to apply the output calibration in myQA Patients, can I apply it after measurement?
Yes, it is possible to apply output calibration in myQA Patients after measurement. Please find the steps below:
Get the Output calibration value you'd like to apply.
You can find it in myQA Platform - Equipment setup - select your 2D array - Calibrations - Edit - select the Output calibration - View
Go back to myQA Patients, click the Tools bar on the right side of the image.
In page Signal - Assign absolute dose, here you can decide how to convert measured cnt to cGy.
For example, in Step 1 the factor is 0,00304cGy/cnt, that means 1cnt=0,00304cGy.
The column allows max 3 digits after the decimal point, we can say 100cnt=0,304cGy.
Click Apply. You will have the measurement in cGy.