What's new in myQA Accept 9.0 SWP-2022-0006 (22.214.171.124)?
myQA Accept 9.0 Patch 2022-0006 fixes some bugs of prior versions further increasing myQA Accept's usability, quality, and efficiency.
Bugs fixed with this version
- 3D DICOM Dose Import
A possible error that might occur while extracting profiles from 3D DICOM dose cubes is the flipping of profiles. This could happen with 3D DICOM dose cubes produced with the latest version of Eclipse/Aria (15.7). Another possible error is the creation of artefacts on the extracted profiles and PDDs.
With this patch version both bugs have been fixed and the import of 3D DICOM dose distributions and extraction of PDDs and profiles works perfectly fine with the latest version of Eclipse/Aria.
- Blue Phantom PT
When measuring with a Blue Phantom PT sometimes it could happen that the phantom is not properly displayed in the control panel, e.g. when using the “GoTo” function. This bug has been fixed and the Blue Phantom PT is now properly displayed in the controller panel.
- Export and import of scan regions settings for PDDs
When exporting and importing scan regions settings for PDDs from one installation into another parameters were changed during the import. This bug has been fixed with the software patch and the scan parameters remain untouched.