How to convert Electron PDI to PDD by applying the correct Protocol for the Perturbation effect in myQA Accept Software?
We can perform the following steps for that.
In myQA Accept software for TG25 and TG51 calculation is not correct for Electron PDI to PDD conversion for PPC chambers, in other words, the code does not know if it is CC or PPC chamber. So one can use this Protocol by deselecting Perturbation correction in TG25/TG51 or another Protocol for this conversion like TRS 398 so that there is a correct R50 value for PPC.
If TG25 or TG51 protocol is being used during Electron PDI to PDD conversion and to get the correct value of R50 then, Go to Data Handling<Ionization to Dose conversion<TG25 protocol<Deselect Perturbation correction as shown below:
Further, this bug will be solved in the next version of myQA Accept until this article can be used.