How to measure point dose in myQA Accept?
The point dose measurement is to measure an absolute dose at a given point on the central axis of the chamber.
Calibration Temperature and Pressure are applied as per calibration certificates.
Note: The point of measurement for cylindrical chambers, on the central axis at the centre of the cavity volume and for plane-parallel chambers, on the inner surface of the window at its centre.
In Queue item settings select Absolute Point in Scan Type.
In Basic settings, type in the Measurement time and the coordinates of the measured point.
In the Extended settings, type in a value in No. of measurements. The average of these
measurements are used for calculation.