How to connect hardware (eg. MatriXX Evolution, Startrack) through different subnetworks in myQA?
Please follow the below steps to connect hardware through different subnetworks in myQA:
- myQA version 2.9|2017-002 or newer is installed
- In the workspace myQA Platform > Equipment Setup at least one clinic and one machine exist.
Set IP address and Network mask for the hardware, please find detailed steps here: How do I configure MatriXX for different subnetworks?
Close myQA, open C:\Program Files (x86)\IBA Dosimetry\myQA\IBADos.CSP.Run.Shell.exe.config with Notepad. (If myQA is installed in C disk).
Find the key “MultiNetworkMode” and set the value to “1”. Then Save the file.
Run myQA - select the clinic/department/room where you want to add the hardware to - click New Detector button.
The hardware will not appear online, thus please input Detector properties manually and then click Save button to apply changes.
Now you can go to myQA Patient/Machines/Fasttrack to test the connection.