How to get started with MatriXX Resolution?
MatriXX Resolution is a 2D dose measurement detector that supports WIFI connection and battery power supply. There are 1521 ion chambers inside MatriXX Resolution. The chamber distance is 6,5mm. The active area is 25.3cmx25.3cm. Here are the steps to get started with MatriXX Resolution
Unpack MatriXX Resolution
Power on MatriXX Resolution
Set up network configuration. You can find detailed steps here: How to configure the MatriXX RESOLUTION network settings?
Setup MatriXX Resolution on the treatment couch for measurement. We suggest using our Miniphantom R here. We also provide different inserts, please see What inserts are avaible for miniphantom R?
Unpack Miniphantom R and set it up together with MatriXX Resolution.
In case you want to mount MatriXX Resolution with a Gantry holder, please follow the below video.
Please note that the SSD is a little bit different from the standard SSD:
ØLong holder SDDMXX_Res = 101.37cm and not 100cm
ØShort holder SDDMXX_Res = 77.5cm and not 76.2cm
Mount gantry angle sensor GS+. (optional)
Now it is ready for myQA connection and measurement.
Here is a video of typical clinical routine workflow